1 Department of Micro- and Nanotechnology, Technical University of Denmark2 Nanoprobes, Department of Micro- and Nanotechnology, Technical University of Denmark3 unknown4 Center for Intelligent Drug Delivery and Sensing Using Microcontainers and Nanomechanics, Center, Technical University of Denmark
In this work, we present a compact, simple and efficient platform for Micro-electromechanical systems (MEMS)/Nano-electromechanical systems (NEMS) characterization. In this platform, a CCD camera is combined with a DVD optical pick-up unit (OPU). The CCD camera captures optical image of MEMS/NEMS samples and detection laser spot, which makes laser alignment on measurement target easier. The DVD OPU is used for detection of resonant frequency measurements of the samples. Working bandwidth and noise level of the OPU are 100 MHz and 1.3 pmHz"2, respectively. Furthermore, the OPU has a laser spot size of 560 run (full width at half maximum, FWHM), which is capable of measuring cantilevers and strings with sub-micron width. A homemade nano-scale resolution X-Y-Z positioner with working distances of 12, 12, 5 mm is responsible for laser-sample alignment. Both thermal and excited resonant frequencies of MEMS/NEMS cantilevers and strings are characterized.
Nanotechnology 2013: Advanced Materials, Cnts, Particles, Films and Composites: Proceedings, 2013, p. 485-488
Alignment; CCD cameras; Data storage equipment; Exhibitions; Fluidics; Geometrical optics; MEMS; Nanocantilevers; Nanotechnology; Natural frequencies; NEMS; Videodisks; Optical data processing
Main Research Area:
2013 NSTI Nanotechnology Conference and Expo, 2013